Good morning at all .
Βecause i had some false rain readings at last days , i replaced the reed switch from my rain gauge la-crosse 2300 ,the gold plating at reed switch maybe falling off and causing a trembling contact,for this reason i had sometimes false rain readings at days without clouds in the sky.
And until now i do not have false rain readings.
Here i have the project about it.
Zum Glück hatte ich einen solchen Fehler noch nicht.
Translation for Dimitris:
What a luck that I didn't have that problem yet.
Thank you for the translation Epfenbachwetter .Yes if some German weather station operator La-crosse 2300-2305-2315--2307-2317-2350-2355-2308-3600-3610 he have problem about like me ,he can to replace very easy and very little cost 1 euro here to Greece the reed switch.